Company: Medica Corp.
Date of Enforcement Report: 9/5/01
RECALL NUMBER, PRODUCT AND CODE:
Z-0951-1 EasyBlood Gas Analyzer.
PCO2 and PO2 values are incorrect if displayed in SI units instead of mmHg.
EasyBlood Gas Analyzers with Software version below M4.20.
Medica Corp., Bedford, MA.
Manufacturer, by fax on 8/15/00.
FIRM INITIATED RECALL:
Malaysia, Russia, Hong Kong, Switzerland, Argentina, Czech Republic, Canada, Hungary, Chile Finland, Turkey, Italy, South Africa, Korea, Pakistan, Greece, Portugal, Mexico, Venezuela, Philippines, Bangladesh.