Company: Draeger Medical, Inc.
Date of Enforcement Report 12/30/2015
Optional PS500 Power Supply Unit for the Evita V500 Ventilator and Babylog VN500 Ventilator. Babylog VN500 is a ventilation unit intended for the ventilation of neonatal and pediatric patients. Recall Number Z-0436-2016
The firm became aware of cases in which the battery run times of the optional PS 500 power supply unit with the Infinity Workstation Critical Care (Evita Infinity V 500) were unexpectedly short due to the design of the charging algorithm in the current software. Devices used for patient transport will be a priority.
Draeger Medical, Inc., Telford, PA on 12/1/2015. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE