Company: Sysmex America, Inc.
Date of Enforcement Report: 4/22/2020
The CF-70 instrument (product code: LXG; Regulation Number: 862.2050) receives magazines containing stained blood smears from SP-50, selects only smears that require analysis, and transports them to the DI-60 for further analysis. After completing the analysis on DI-60 the smears are placed back in magazines.
Recall Number: Z-1703-2020
Software mismatch-When the software versions between the SP-50 and CF-70 are not matched, and an error condition occurs requiring the operator to open the cover door of the CF-70 to remove slides or slide magazines from the CF-70, the operation of the CF-70 does not halt.
Sysmex America, Inc. on 3/9/2020. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE