Company: Siemens Medical Solutions USA, Inc
Date of Enforcement Report: 3/24/2021
Software versions syngo.CT VA30A_SP2 or syngo.CT VA30A_SP2a in the Computed tomography systems as follows: SOMATOM go.Now (Model 11061610)SOMATOM go.Up (Models 11061620 & 11061628) SOMATOM go.All (Model 11061630) SOMATOM go.Top (Models 11061640 & 11061648) SOMATOM go.Sim (Model 11061660) SOMATOM go.Open Pro (Model 11061670) SOMATOM X.cite (Model 11330001)
Recall Number: Z-1253-2021
SOMATOM systems-issues in with software syngo.CT VA30A_SP2 or syngo.CT VA30A_SP2a, may result in sporadic problems causing scanning workflow interruptions and unexpected user notifications. Delay in diagnosis or patient rescan may occur. Sporadic software errors during interventional workflows may also result
Siemens Medical Solutions USA, Inc on 3/3/2021. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
190 units US