Company: Siemens Medical Solutions USA, Inc
Date of Enforcement Report: 4/28/2021
Sensis/ Sensis Vibe Hemo systems with VD12A software as follows: Dedicated SIS Server- 6648153 Sensis Post-Processing Workstation- 6648161 Sensis- 10764561 VM Virtual Server- 10765502 Sensis High-End Servers- 10910620 Sensis Vibe Hemo- 11007641 Sensis Vibe Combo- 11007642 Sensis Vibe is a recording and procedure data management system for interventional cardiology, interventional radiology, surgical procedures performed in a hybrid OR and electrophysiology.
Recall Number: Z-1434-2021
Due to the configuration of certain Windows Service Permissions within the operating systems of the Sensis/ Sensis Vibe computer, there is a risk for exposure of sensitive information, manipulation of data, or Denial of Service attacks and could result in incorrect diagnostic or therapeutic decisions.
Siemens Medical Solutions USA, Inc on 3/4/2021. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE