Company: Biocare Medical, LLC
Date of Enforcement Report: 4/7/2021
BIOCARE Medical , IntelliPath FLX, Automated Staining Instrument, Ref: ” Model Number:IPS0001US and IPS0001INTL containing software version 184.108.40.206
Recall Number: Z-1333-2021
Their is a potential that the automated staining instrument with software version 220.127.116.11 may move with random speeds in random direction across its range of motion. This could result in the instrument hitting end stops or running into other components on the working deck. This could also result in wash buffer being dispensed on random locations.
Biocare Medical, LLC on 2/25/2021. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE
U.S. Nationwide and International