Company: Siemens Medical Solutions USA, Inc
Date of Enforcement Report: 7/14/2021
ARTIS pheno- angiography systems developed for single and biplane diagnostic imaging and interventional procedures. Model: 10849000
Recall Number: Z-2029-2021
Software error VE10B -In case of a fault within the drivetrain of an axis and while performing a stepped peripheral imaging (PERISTEPPING/PERIVISION), the C-arm may leave its intended travel path by up to 22 cm (9″) result in canceling clinical treatment or to continue treatment on an alternative system
Siemens Medical Solutions USA, Inc. on 5/27/2021. Voluntary: Firm Initiated recall is ongoing.
VOLUME OF PRODUCT IN COMMERCE